The Sechrist IV-200 Infant/Pediatric Ventilator is a pressure-limited, time cycled, continuous flow ventilator. Manual breaths and CPAP is also available. The Sechrist IV-200 control circuitry can be operated with 100-240 volts AC, 50/60 Hz. It can also be operated from a 12 volt DC power supply using the accessory power cable provided. This ventilator requires pneumatic power and operates on external compressed air and oxygen sources from 30-60 psi. The gas mixing blender and flowmeter are enclosed within the ventilator housing.
The Sechrist IV-200 ventilator is capable of meeting the most demanding requirements of the NICU or PICU around the world. The built-in displays provide easy-to-read continuous ventilation parameter information.
The alarm package and modes of ventilation including CPAP are similar to the IV100B infant ventilator system. This ventilator can adequately ventilate newborns to pediatric patients with precise control of oxygen concentration, at respiratory rates from 1 to 150 BPM. To prevent inadvertent PEEP when higher flow rates are used, a special pediatric exhalation valve block and pediatric circuit are available.
Ventilation Application: Neonatal, Pediatric
Modes of Ventilation:
Flow: 0 to 32 LPM; Flush 40 LPM
Oxygen Percent: 21% to 100%
Inspiratory Time: 0.10 to 2.90 sec.
Expiratory Time: 0.30 to 60.0 sec.
Inspiratory Pressure: 5 to 70 cm H2O
Expiratory Pressure: -2 to 20 cm H2O
Test: Checks microprossor function, displays and alarm function. Waveform Control. Allows the user to vary the proximal airway pressure waveform.
Time Preset: While in the CPAP mode, this feature allows the user to preview and select the desired Inspiratory and Expiratory Times before switching to the VENT mode.
Safety Pressure Relief Valve (Adjustable): 15 to 85 cm H2O
3 to 60 sec. Sets the time delay before the LO pressure alarm is audibly and visually activated.
Alarm Mute: 30 � 5 sec.
Alarm Set Arrows: High Inspiratory Pressure Alarm Limit. Low Inspiratory Pressure Alarm Limit.
Select (Reset) Button: Allows the user to select either the LO or HI pressure alarm limit setting. This button is also used to cancel the HI limit alarm when it is activated.
High (HI) Pressure Limit
Low (LO) Pressure Limit
Low Airway Pressure
Source Gas Failure
Power Failure (Battery Backup)
Airway Pressure (to 80 cm H2O)
High Pressure Alarm
Low Pressure Alarm
Mean Airway Pressure
Inverse I:E Ratio
High and Low Pressure Alarm
Limit Set Points
Pre-set: Safety Auto Lock-Out Circuit (Inspiratory Phase terminated at 4 sec. Even if microprossor fails.
Pneumatic Requirements: Medical Grade Air and Oxygen: 30 to 60 psi; differential within 20 psi.
117 VAC 50/60 Hz
Overseas voltage requirements available)
Circuit Breaker Protected
D.C. Power Requirements
12 VDC 1.2 amps
Height: 16.25 in (41.27 cm)
Width: 12 in (30.48 cm)
Depth: 9.5 in (24.13 cm)
Net: 27 lbs. (12.2 Kg)
Shipping: 35 lbs. (15.9 Kg)
3-Wire SJT Power Cord. Hospital-Grade Plug, current leakage less than 20 microamps when tested per UL 544. Auxiliary Flowmeter Port.